Range & Resolution
±1.5%F | ±0.3°C
15ft (5 meters)
The EC-1 sensor is meant to be fully submerged in your tank/reservoir. You can use the included mounting bracket for mounting the sensor, or you can use the 3/4" NPT threading for inline installation. When mounting make sure the positioning allows the tip of the sensor to remain fully submerged and in a spot where it won't get knocked around.
Use and maintenance
Your sensor(s) will need periodic cleaning and calibration. The maintenance cycle can vary depending on your environment. To ensure you receive a long life from your sensor, please follow proper use/maintenance instructions.
Remove plastic cap from the tip of the sensor and run under fresh tap water (Do not use distilled, de-ionized, or reverse osmosis water).
Fill a small container with clean tap water.
Add a small amount of probe cleaner or mild dishwashing liquid.
Gently stir sensor in solution. *Make sure you don't knock/bump the sensor off the sides of container as this can cause damage*
Incorrect usage of this sensor voids our warranty.
Part 1 - Zero Point Calibration
Wipe Your Sensor Dry
Suspend the sensor vertically in the air
Wait for your readings to stabilize
Press the “Calibrate” button.
Part 2 - Slope Calibration
Fill a non-metallic container with calibration solution
Enter calibration solution value into input prompt
Swirl sensor in solution without touching the walls or bottom
Suspend sensor in solution 1 inch (2.5cm) from the walls and bottom
Wait for the reading to stabilize
Press the "Calibrate" button
If you require additional support with calibration, reach out to us at firstname.lastname@example.org
Frequently Asked Questions (FAQ)
Can my sensor remain fully submerged? Yes! Your sensor can remain fully submerged in your tank/reservoir.
Running into issues? Reach out to us at email@example.com or give us a call at 888-537-8573 during our business hours (M-F 9a-4p PT) so we can take care of you!